Validation of a magnetic resonance imaging guided stereotactic access to the ovine brainstem
© Staudacher et al.; licensee BioMed Central Ltd. 2014
Received: 24 February 2014
Accepted: 10 September 2014
Published: 22 September 2014
Anatomical differences between humans and domestic mammals preclude the use of reported stereotactic approaches to the brainstem in animals. In animals, brainstem biopsies are required both for histopathological diagnosis of neurological disorders and for research purposes. Sheep are used as a translational model for various types of brain disease and therefore a species-specific approach needs to be developed. The aim of the present study was to establish a minimally invasive, accurate and reproducible stereotactic approach to the brainstem of sheep, using the magnetic resonance imaging guided BrainsightTM frameless stereotactic system.
A transoccipital transcerebellar approach with an entry point in the occipital bone above the vermis between the transverse sinus and the external occipital protuberance was chosen. This approach provided access to the target site in all heads. The overall mean needle placement error was 1.85 ± 1.22 mm.
The developed transoccipital transcerebellar route is short, provides accurate access to the ovine caudal cranial fossa and is a promising approach to be further assessed in live animals.
Currently, stereotactic brain biopsy is the least invasive method to obtain brain tissue -, especially from lesions that are deep-seated ,- or located in vitally important intracranial regions such as the brainstem ,,-. Three different approaches to the brainstem have been described in people: the transtentorial, transfrontal and suboccipital transcerebellar route -. Since traversing the tentorium might cause pain and/or hemorrhage at the pial surfaces of the cerebellum or mesencephalon , and potentially damage vital blood vessels and cranial nerve nuclei, the transtentorial route is no longer used . Depending on the location of the lesions, either a transfrontal ,,- or a suboccipital transcerebellar ,- approach is used. For midbrain lesions, targets located along the midline of the brainstem and for lesions in the caudal part of the medulla oblongata, a transfrontal coronal route is used, whereas for lesions located laterally or within the cerebellum, cerebellar peduncles and pons, a suboccipital transcerebellar route through the middle cerebellar peduncles is preferred -.
In animals, brainstem biopsies are required both for histopathological diagnosis of neurological disorders and for research purposes. Sheep are used as a translational model for various types of brain diseases in humans - and brainstem biopsies are necessary to investigate the neuropathogenesis of listeric rhombencephalitis, the most frequent central nervous system disease of ruminants. However, the anatomical differences between species do not allow methods developed in humans to be transferred directly to the sheep.
The intraparenchymal transcerebellar approach to the brainstem is inadequate in domestic mammals because of the lateral position of the cerebellar peduncles . Moreover, the caudal contour of the cerebellum is covered by the squama occipitalis, and the attachment of nuchal muscles precludes a suboccipital transcerebellar approach (Figure 1).
Consequently, stereotactic approaches to the brainstem used in human medicine cannot be applied to sheep. In veterinary medicine, brain targets located within the caudal cranial fossa were rarely addressed stereotactically ,, but the employed approach was not mentioned. In the present study an applicable transoccipital transcerebellar magnetic resonance imaging guided stereotactic approach to the brainstem of sheep is described and its target accuracy was determined using the modified BrainsightTM stereotactic system.
The mean needle placement error for the midbrain (n = 6), pons (n = 6) and obex (n = 6) targets was 1.77 ± 1.47, 2.48 ± 1.16 and 1.28 ± 0.83 mm, respectively. The overall mean needle placement error for all target sites (n = 18) was 1.85 ± 1.22 mm. The mean target depth for the midbrain, the pons and the obex targets was 36.9 ± 2.36, 33.18 ± 0.82 and 29.6 ± 1.50 mm, respectively. There was no statistically significant relationship between needle placement error and target depth (P = 0.28).
Macroscopic evaluation revealed that toluidine stains were visible in the region of sixteen of the eighteen targeted sites. In two brains, a blue stained margin was observed at the edge of the obex. The position of the toluidine stain was at the target site in thirteen sheep heads. The dot was displaced in three brainstems. It was found rostrally to one target in the left midbrain, medially to one target in the right pons and lateroventrally to a destination in the left pons.
The stereotactic transoccipital transcerebellar approach to the ovine brainstem used in this study was applicable in all the heads used. The overall mean needle placement error in the brainstem of 1.85 ± 1.22 mm is comparable to previous results of the magnetic resonance imaging guided BrainsightTM stereotactic system in targets in the canine rostral and middle cranial fossa (mean needle placement error of 1.79 ± 0.87 mm) . Reported mean needle placement errors of CT-guided stereotactic systems used in veterinary medicine are larger for targets within the rostral, middle and/or caudal cranial fossa ,,, and only slightly smaller (1.7 ± 1.6 mm) for targets exclusively located in the rostral cranial fossa .
The obtained error was also smaller in comparison with the in vivo accuracy of frameless MRI guided stereotactic brainstem biopsy sampling in human medicine ,. Frameless systems have replaced the stereotactic frame with a method of registration that relies on anatomic landmarks - such as nose, eyes and ears - or artificial markers, called `fiducials'. The latter are attached to the patient's head before the brain scan, and a three-dimensional digitizer matches them to the corresponding points in the image. Frameless systems provide a wide range of motion for the instrument guidance arm .
We consider the transcerebellar approach with its entry point in the occipital bone above the vermis between the transverse sinus and the external occipital protuberance to be the only one to be safely applied to the whole ovine brainstem (Figure 1). The brain surface through which a trajectory to the brainstem can be placed is rostroventrally confined by the large frontal sinuses -. As in people, the trajectory should avoid the membranous tentorium of the cerebellum . Therefore, the transfrontal approach, which permits access to all divisions of the brainstem ,, has limited value in domestic mammals and the transcerebellar route is more promising. However, a transcerebellar trajectory may not enter through the squama occipitalis, the part of the occipital bone situated caudally to the cerebellum, because the required shallow angle between bone and drill bit would result in slippage of the drill bit . Furthermore, the occipital squama of sheep is covered by a considerable amount of muscles, whose dissection is known to cause massive postoperative wound pain in people ,,. Consequently, the entry point has to be placed rostral to or on the external occipital protuberance (Figures 1, 3).
Attention must be paid to the vasculature, notably the transverse sinus ,,,,, the dorsal cerebellar veins and dorsal rami of the caudal cerebellar arteries between the vermis and the cerebellar hemispheres , which limit the area allowing the caudal cranial fossa to be accessed cranially and laterally, respectively.
Using this approach, there is no lateral restriction to the described trajectory. Another important advantage is the shortness of the trajectory ,,,,,,,, which results in minimized tissue trauma as well as increased accuracy ,.
Disadvantages of the human suboccipital transcerebellar compared to the transfrontal approach such as the need for general anesthesia , prone positioning , and considerable muscle dissection ,, are irrelevant in veterinary medicine: all the patients need to be subjected to general anesthesia, thus rendering participation in an intraoperative neurological examination impossible ,. The tetrapod anatomy suggests prone positioning during the MRI examination and the interventional procedure, which is a comfortable operating position for the surgeon  and complies with the physiological patient composure so that brain shift is a minor concern . In this cadaver head study, loss of CSF and elasticity of the brain parenchyma as well as loss of continuity of the brainstem with the spinal cord may have induced brain shift ,,. On the other hand, lack of parenchymal excursions in temporal synchrony with systole  could have led to underestimation of targeting error, but this impact can only be assessed in a clinical setting.
Other causes of needle placement error might also have occurred during target registration, fiducial registration, and target positioning . In order to minimize these errors, a rigid head fixation was ensured and the registration was checked after each step that could have caused slippage of the head in the clamp or movement of the freeguide arm . In accordance with published recommendations, the fiducial markers implanted into the bone were carefully monitored throughout the whole procedure by the same previously trained person ,,.
As judged from macroscopic evaluation, the toluidine dot was displaced in three specimens. In all three specimens, the actual needle placement error was higher than the mean needle placement error (4.35 mm, 3.15 mm and 2.01 mm). Nevertheless, in some objects with a large needle placement error (4 mm, 3.23 mm), the toluidine stain was not judged to be off target macroscopically, possibly due to the fact that an error with a large deviation in a single plane is more striking than an error with a small deviation in all three planes. In two brains, in which no toluidine stain but a blue margin of contrast medium at the edge of the obex was detected, the contrast medium presumably leaked when the injection needle pierced the meninges, a known drawback of stereotactic drug delivery ,, which is probably of minor importance in stereotactic brain biopsy sampling. Access to the rostral mesencephalon is limited by the tentorium and the external occipital protuberance, which prohibits more caudal tilting of the trajectory. Therefore, in our experience, the crura cerebri are the most rostral area, which can be accessed through the cerebellum. The caudal restriction of the transcerebellar approach is given by the caudal border of the cerebellum, making the obex the most caudal area which can be reached. Additionally the transverse sinus prevents further rostral tilting of the trajectory.
Consequently, the aforementioned transoccipital transcerebellar approach gives access to all structures within the ovine caudal cranial fossa including the cerebellum, cerebellar peduncles and lateral regions of the brainstem.
Intracranial hemorrhage and postoperative neurological deficits are a major concern in human stereotactic brainstem biopsy ,,, and could also arise in sheep. Although the herein developed approach to the ovine brainstem avoids the transverse sinus, the dorsal cerebellar veins, the dorsal rami of the caudal cerebellar arteries between the vermis and the cerebellar hemispheres and cranial nerves within the brainstem, the risk of bleeding from smaller vessels as well as the occurrence of neurological deficits remains to be systematically assessed in living sheep.
In contrast to sheep, the canine skull has more prominent bony crests in the occipital area. Depending on dog breed and size, this prohibits an entry point in the midline of the occipital bone. However, the anatomy of the ovine, canine and feline head is otherwise similar so that the herein described approach to the brainstem can theoretically be translated to dogs and cats . The authors have already employed a transoccipital transcerebellar approach in canine cadavers.
The study proved a stereotactic transoccipital transcerebellar approach to be suitable to access targets along the whole axis of the ovine brainstem with good accuracy and is currently used to sample the brainstem in live sheep. Possibly associated complication rates and the application of the access in other species can now be assessed in further studies.
Eighteen one-year-old healthy sheep of different breeds (Swiss White Alpine Sheep (n = 3), Black- and Brownheaded Mutton (n = 3 and n = 12, respectively) were slaughtered in the context of food production. The study was performed in agreement with the local ethic regulations (Swiss Veterinary Service, Office of Agriculture and Nature (LANAT)). The heads were disarticulated at the atlantooccipital junction and stored at + 6°C within 48 hours upon arrival until use. Mean weight of the heads was 2.96 ± 0.91 kg (1.92-4.62 kg). The stereotactic procedures were performed using the BrainsightTM frameless stereotactic system (Rogue Research Inc., Montreal, Canada)  with a modified bone implanted fiducial marker system ,,,. After clipping of the coat and placement of the sheep head in the C-clamp, the fiducial marker post was fixed on the frontal bones caudal to one of the zygomatic processes to ensure that the registration markers were in different planes and not obstructing the surgical site. Hence, the distance from the centroid of all fiducial markers to the planned target was small ,, and suitable bone thickness for post fixation was ensured. The implant post was attached with at least three 8 mm-ceramic screws. A fiducial array hub with five fiducial markers was screwed onto the post.
The cadaver head was scanned in a 1.0 Tesla MRI system (Philips Panorama HFO, Philips System, Best, The Netherlands) in prone position using a head coil. A T1-weighted gradient echo 3D sequence was acquired using the following parameters: TR = 25 ms, TE = 6.9 ms, flip angle = 30°, Number of Signal Averages = 2, slice thickness = 1.8 mm without interslice gap. The field of view was adjusted according to skull size and position of fiducial markers.
Three different target sites within the brainstem were determined: ventral part of the midbrain directly adjacent to the pons, pons at the level of the emergence of the facial nerve or obex, on the right or left side, respectively (Figure 2). This resulted in three cadaver heads per target site. For each target, coordinates (X, Y, Z) were read out using the BrainsightTM neuronavigation software. The trajectory for all the targets was planned via a transoccipital transcerebellar access (Figures 3, 4).
Following acquisition of the MRI, the sheep head with the fiducial marker system was again fixed in the surgical headclamp in prone position using four skull screws. The open side of the surgical C-clamp was directed caudally. The two skull screws at the caudal end of the clamp were placed in the temporal fossa. The rostral skull screws made contact with the nasal bone.
The subject to image-registration was performed  with the Polaris® optical position sensor placed rostrally to the cadaver head and checked directly before and after drilling and after removal of the manual ruler guide from the instrument sleeve. The neuronavigation pointer was inserted into the instrument sleeve of the articulated arm. The entry point was kept as perpendicular to the skull surface as possible to prevent slippage of the drill bit. Tight locking of the articulated arm ,, correct use of the stabilization pin and good contact between the drill guide tube and the skull  were ensured. Furthermore the instrument receptacle of the articulated arm was fixed manually during the drilling of a 5 mm burr hole to prevent residual movement of the assembly.
The neuronavigation pointer was inserted in the instrument sleeve and lowered down to a zeroing platform. The distance from the zeroing platform to target was determined by the software . The manual ruler guide with a 10 μl syringe and attached 26-gauge, 6-inches needle (Hamilton Company, Reno NV/Bonaduz, Switzerland) filled with the contrast solution and straightened by a guiding cannula was then placed in the instrument sleeve of the articulated arm. The needle was lowered manually to the zeroing platform, the manual ruler guide was set to zero and after removal of the platform the needle was lowered to target.
Ten minutes after the needle reached the predetermined target depth, 0.5 μl of the contrast solution was injected. This contrast solution was made by adding 0.4 ml of gadodiamide (Omniscan®, GE Healthcare Inc., Glattbrugg, Switzerland) and 0.05 g of toluidine blue to 100 ml of 0.9% NaCl. The needle was kept in place for 5 min after the injection to prevent contrast medium leakage along the needle track. Immediately following the contrast injection, a second MRI study was performed using the same parameters as before. These images were uploaded and saved to the neuronavigation computer later on .
To determine the mean needle placement error, the postoperative MR images were registered to the preoperative ones using the Image Feature of the BrainsightTM neuronavigation software. The preoperative MRI served as the reference so that a common coordinate space was used between the preoperative and postoperative MRI. The planned injection site represented target A and the center of the gadodiamide deposition target A' . Coordinates of A' (X', Y', Z') were read off. In two cases, the contrast bloom was not clearly visible but a gas bubble was depicted at the tip of the needle track on the MRI. The center of these gas bubbles was taken as the target. The precision of the system in bringing the needle to target (needle placement error) was calculated for each target site using the formula: Error = √[(X-X')2 + (Y-Y')2 + (Z-Z')2]. Mean needle placement error and standard deviation (SD) were then assessed based on all the target sites. Linear regression was used to evaluate the relationship between needle placement error and target depth in the brain. P value < 0.05 was considered significant .
All authors participated in the general design and conception of the study. MS and AS developed the transoccipital transcerebellar stereotactic approach to the brainstem of sheep. AS carried out the stereotactic procedures, acquired magnetic resonance images, determined needle placement error, read off target depths and collected, fixed and sliced the brains. AO conducted the contrast stain evaluation. DG performed the statistical analysis. AS and DG drafted the manuscript. AS, AO, MS and DG revised the manuscript and gave final approval of the version to be published. All authors agree to be accountable for all aspects of the work. All authors read and approved the final manuscript.
The authors thank Kaspar Stucki for statistical support and Franziska Hostettler, Brigitte Löffel, Peter Ruprecht and the technical service of the Vetsuisse-faculty Bern for their technical assistance. There was no funding to support the study.
- Boviatsis EJ, Voumvourakis K, Goutas N, Kazdaglis K, Kittas C, Kelekis DA: Stereotactic biopsy of brain stem lesions. Minim Invasive Neurosurg. 2001, 44 (4): 226-229. 10.1055/s-2001-19930.PubMedGoogle Scholar
- Boviatsis EJ, Kouyialis AT, Stranjalis G, Korfias S, Sakas DE: CT-guided stereotactic biopsies of brain stem lesions: personal experience and literature review. Neurol Sci. 2003, 24 (3): 97-102. 10.1007/s10072-003-0093-3.PubMedGoogle Scholar
- Chico-Ponce de León F, Perezpena-Diazconti M, Castro-Sierra E, Guerrero-Jazo FJ, Gordillo-Dominguez LF, Gutierrez-Guerra R, Salamanca T, Sosa-Sainz G, Santana-Montero BL: DeMontesinos-sampedro A: stereotactically-guided biopsies of brainstem tumors. Childs Nerv Syst. 2003, 19 (5-6): 305-310. 10.1007/s00381-003-0737-x.PubMedGoogle Scholar
- Kratimenos GP, Nouby RM, Bradford R, Pell MF, Thomas DG: Image directed stereotactic surgery for brain stem lesions. Acta Neurochir (Wien). 1992, 116 (2-4): 164-170. 10.1007/BF01540871.Google Scholar
- Rajshekhar V: Current status of stereotactic biopsy. Stereotact Funct Neurosurg. 2001, 76 (3-4): 137-139. 10.1159/000066709.PubMedGoogle Scholar
- Sawin PD, Hitchon PW, Follett KA, Torner JC: Computed imaging-assisted stereotactic brain biopsy: a risk analysis of 225 consecutive cases. Surg Neurol. 1998, 49 (6): 640-649. 10.1016/S0090-3019(97)00435-7.PubMedGoogle Scholar
- Giese H, Hoffmann KT, Winkelmann A, Stockhammer F, Jallo GI, Thomale UW: Precision of navigated stereotactic probe implantation into the brainstem. J Neurosurg Pediatr. 2010, 5 (4): 350-359. 10.3171/2009.10.PEDS09292.PubMedGoogle Scholar
- Platt SR, Olby NJ: BSAVA Manual of Canine and Feline Neurology. 3rd edition. Gloucester, United Kingdom: British Small Animal Veterinary Association; 2004.Google Scholar
- Hood TW, Gebarski SS, McKeever PE, Venes JL: Stereotaxic biopsy of intrinsic lesions of the brain stem. J Neurosurg. 1986, 65 (2): 172-176. 10.3171/jns.1986.65.2.0172.PubMedGoogle Scholar
- McGirt MJ, Woodworth GF, Coon AL, Frazier JM, Amundson E, Garonzik I, Olivi A, Weingart JD: Independent predictors of morbidity after image-guided stereotactic brain biopsy: a risk assessment of 270 cases. J Neurosurg. 2005, 102 (5): 897-901. 10.3171/jns.2005.102.5.0897.PubMedGoogle Scholar
- Amundson EW, McGirt MJ, Olivi A: A contralateral, transfrontal, extraventricular approach to stereotactic brainstem biopsy procedures. Technical note. J Neurosurg. 2005, 102 (3): 565-570. 10.3171/jns.2005.102.3.0565.PubMedGoogle Scholar
- Neal JH, Van Norman AS: Transcerebellar biopsy of posterior fossa lesions using the leksell gamma model stereotactic frame. Neurosurgery. 1993, 32 (3): 473-474. 10.1227/00006123-199303000-00024. discussion 474-475PubMedGoogle Scholar
- Sanai N, Wachhorst SP, Gupta NM, McDermott MW: Transcerebellar stereotactic biopsy for lesions of the brainstem and peduncles under local anesthesia. Neurosurgery. 2008, 63 (3): 460-466. 10.1227/01.NEU.0000324731.68843.74. discussion 466-468PubMedGoogle Scholar
- Cartmill M, Punt J: Diffuse brain stem glioma. A review of stereotactic biopsies. Childs Nerv Syst. 1999, 15 (5): 235-237. 10.1007/s003810050379. discussion 238PubMedGoogle Scholar
- Giunta F, Marini G, Grasso G, Zorzi F: Brain stem expansive lesions: stereotactic biopsy for a better therapeutic approach. Acta Neurochir Suppl (Wien). 1988, 42: 182-186.Google Scholar
- Rajshekhar V, Moorthy RK: Status of stereotactic biopsy in children with brain stem masses: insights from a series of 106 patients. Stereotact Funct Neurosurg. 2010, 88 (6): 360-366. 10.1159/000319044.PubMedGoogle Scholar
- Abernathey CD, Camacho A, Kelly PJ: Stereotaxic suboccipital transcerebellar biopsy of pontine mass lesions. J Neurosurg. 1989, 70 (2): 195-200. 10.3171/jns.1989.70.2.0195.PubMedGoogle Scholar
- Gonçalves-Ferreira A: Stereotactic anatomy of the posterior cranial fossa. A study of the transcerebellar approach to the brainstem. Acta Neurochir (Wien). 1991, 113 (3-4): 149-165. 10.1007/BF01403201.Google Scholar
- Guthrie BL, Steinberg GK, Adler JR: Posterior fossa stereotaxic biopsy using the brown-Roberts-wells stereotaxic system. Technical note. J Neurosurg. 1989, 70 (4): 649-652. 10.3171/jns.1989.70.4.0649.PubMedGoogle Scholar
- Pérez-Gómez JL, Rodriguez-Alvarez CA, Marhx-Bracho A, Rueda-Franco F: Stereotactic biopsy for brainstem tumors in pediatric patients. Childs Nerv Syst. 2010, 26 (1): 29-34. 10.1007/s00381-009-1000-x.PubMedGoogle Scholar
- Coffey RJ, Lunsford LD: Stereotactic surgery for mass lesions of the midbrain and pons. Neurosurgery. 1985, 17 (1): 12-18. 10.1227/00006123-198507000-00003.PubMedGoogle Scholar
- Dellaretti M, Reyns N, Touzet G, Dubois F, Gusmao S, Pereira JL, Blond S: Stereotactic biopsy for brainstem tumors: comparison of transcerebellar with transfrontal approach. Stereotact Funct Neurosurg. 2012, 90 (2): 79-83. 10.1159/000335502.PubMedGoogle Scholar
- Gonçalves-Ferreira AJ, Herculano-Carvalho M, Pimentel J: Stereotactic biopsies of focal brainstem lesions. Surg Neurol. 2003, 60 (4): 311-320. 10.1016/S0090-3019(03)00379-3. discussion 320PubMedGoogle Scholar
- Kondziolka D, Lunsford LD: Results and expectations with image-integrated brainstem stereotactic biopsy. Surg Neurol. 1995, 43 (6): 558-562. 10.1016/0090-3019(95)00009-7.PubMedGoogle Scholar
- Stypulkowski PH, Giftakis JE, Billstrom TM: Development of a large animal model for investigation of deep brain stimulation for epilepsy. Stereotact Funct Neurosurg. 2011, 89 (2): 111-122. 10.1159/000323343.PubMedGoogle Scholar
- Oevermann A, Zurbriggen A, Vandevelde M: Rhombencephalitis caused by listeria monocytogenes in humans and ruminants: a zoonosis on the rise?. Interdiscip Perspect Infect Dis. 2010, 2010: 632513.PubMed CentralPubMedGoogle Scholar
- Förschler A: MRI of experimental focal cerebral ischemia in sheep. RöFo. 2007, 179 (H. 5): 516-524. 2007PubMedGoogle Scholar
- Nickel RSA, Seiferle E: Zentralnervensystem. In Lehrbuch der Anatomie der Haustiere. Volume IV, 3rd edition. Edited by Böhme G. Berlin und Hamburg: Parey Buchverlag; 1992:28–228.Google Scholar
- Nickel RSA, Seiferle E: Knochenlehre. In Lehrbuch der Anatomie der Haustiere. Volume I, 6th edition. Edited by Böhme G. Berlin und Hamburg: Parey Buchverlag; 1992.Google Scholar
- Moissonnier P, Bordeau W, Delisle F, Devauchelle P: Accuracy testing of a new stereotactic CT -guided brain biopsy device in the dog. Res Vet Sci. 2000, 68 (3): 243-247. 10.1053/rvsc.1999.0370.PubMedGoogle Scholar
- Koblik PD, Lecouteur RA, Higgins RJ, Fick J, Kortz GD, Sturges BK, Pascoe PJ: Modification and application of a pelorus mark III stereotactic system for CT-guided brain biopsy in 50 dogs. Veterinary Radiol Ultrasound. 1999, 40 (5): 424-433. 10.1111/j.1740-8261.1999.tb00370.x.Google Scholar
- Moissonnier P, Blot S, Devauchelle P, Delisle F, Beuvon F, Boulha L, Colle MA, Lefrancois T: Stereotactic CT-guided brain biopsy in the dog. J Small Anim Pract. 2002, 43 (3): 115-123. 10.1111/j.1748-5827.2002.tb00041.x.PubMedGoogle Scholar
- Troxel MT, Vite CH: CT-guided stereotactic brain biopsy using the kopf stereotactic system. Veterinary Radiol Ultrasound. 2008, 49 (5): 438-443. 10.1111/j.1740-8261.2008.00403.x.Google Scholar
- Kamiryo T, Jackson T, Laws E: A methodology designed to increase accuracy and safety in stereotactic brain surgery. Minim Invasive Neurosurg. 2000, 43 (1): 1-3. 10.1055/s-2000-8811.PubMedGoogle Scholar
- Koblik PD, Le Couteur RA, Higgins RJ, Bollen AW, Vernau KM, Kortz GD, Ilkiw JE: CT- Guided brain biopsy using a modified pelorus mark III stereotactic system: Experience with 50 dogs. Vet Radiol Ultrasound. 1999, 40 (5): 434-440. 10.1111/j.1740-8261.1999.tb00371.x.PubMedGoogle Scholar
- Chen AV, Wininger FA, Frey S, Comeau RM, Bagley RS, Tucker RL, Schneider AR, Gay JM: Description and validation of a magnetic resonance imaging-guided stereotactic brain biopsy device in the dog. Vet Radiol Ultrasound. 2012, 53 (2): 150-156. 10.1111/j.1740-8261.2011.01889.x.PubMedGoogle Scholar
- Taylor AR, Cohen ND, Fletcher S, Griffin JF, Levine JM: Application and machine accuracy of a new frameless computed tomography-guided stereotactic brain biopsy system in dogs. Veter Radiol Ultrasound Off J Am Coll Veter Radiol Int Veter Radiol Assoc. 2013, 54 (4): 332-342. 10.1111/vru.12025.Google Scholar
- Dorward NL, Alberti O, Palmer JD, Kitchen ND, Thomas DG: Accuracy of true frameless stereotaxy: in vivo measurement and laboratory phantom studies. Technical note. J Neurosurg. 1999, 90 (1): 160-168. 10.3171/jns.1999.90.1.0160.PubMedGoogle Scholar
- Paleologos TS, Dorward NL, Wadley JP, Thomas DG: Clinical validation of true frameless stereotactic biopsy: analysis of the first 125 consecutive cases. Neurosurgery. 2001, 49 (4): 830-835. discussion 835-837PubMedGoogle Scholar
- Chen SY, Chen CH, Sun MH, Lee HT, Shen CC: Stereotactic biopsy for brainstem lesion: comparison of approaches and reports of 10 cases. J Chin Med Assoc. 2011, 74 (3): 110-114. 10.1016/j.jcma.2011.01.024.PubMedGoogle Scholar
- Spiegelmann R, Friedman WA: Stereotactic suboccipital transcerebellar biopsy under local anesthesia using the cosman-Roberts-wells frame. Technical note. J Neurosurg. 1991, 75 (3): 486-488. 10.3171/jns.1991.75.3.0486.PubMedGoogle Scholar
- Pincus DW, Richter EO, Yachnis AT, Bennett J, Bhatti MT, Smith A: Brainstem stereotactic biopsy sampling in children. J Neurosurg. 2006, 104 (2 Suppl): 108-114.PubMedGoogle Scholar
- Roujeau T, Machado G, Garnett MR, Miquel C, Puget S, Geoerger B, Grill J, Boddaert N, Di Rocco F, Zerah M, Sainte- Rose C: Stereotactic biopsy of diffuse pontine lesions in children. J Neurosurg. 2007, 107 (1 Suppl): 1-4.PubMedGoogle Scholar
- Zrinzo L, van Hulzen AL, Gorgulho AA, Limousin P, Staal MJ, De Salles AA, Hariz MI: Avoiding the ventricle: a simple step to improve accuracy of anatomical targeting during deep brain stimulation. J Neurosurg. 2009, 110 (6): 1283-1290. 10.3171/2008.12.JNS08885.PubMedGoogle Scholar
- Valdés-Gorcía J, Espinoza-Diaz DM, Paredes-Diaz E: Stereotactic biopsy of brain stem and posterior fossa lesions in children. Acta Neurochir (Wien). 1998, 140 (9): 899-903. 10.1007/s007010050191.Google Scholar
- Barnett GH, Miller DW, Weisenberger J: Frameless stereotaxy with scalp-applied fiducial markers for brain biopsy procedures: experience in 218 cases. J Neurosurg. 1999, 91 (4): 569-576. 10.3171/jns.1999.91.4.0569.PubMedGoogle Scholar
- Wang MY, Maurer CR, Fitzpatrick JM, Maciunas RJ: An automatic technique for finding and localizing externally attached markers in CT and MR volume images of the head. IEEE Trans Biomed Eng. 1996, 43 (6): 627-637. 10.1109/10.495282.PubMedGoogle Scholar
- Widmann G, Stoffner R, Sieb M, Bale R: Target registration and target positioning errors in computer-assisted neurosurgery: proposal for a standardized reporting of error assessment. Int J Med Robot. 2009, 5 (4): 355-365. 10.1002/rcs.271.PubMedGoogle Scholar
- Dickinson PJ, LeCouteur RA, Higgins RJ, Bringas JR, Larson RF, Yamashita Y, Krauze MT, Forsayeth J, Noble CO, Drummond DC, Kirpotin DB, Park JW, Berger MS, Bankiewicz KS: Canine spontaneous glioma: a translational model system for convection-enhanced delivery. Neuro Oncol. 2010, 12 (9): 928-940. 10.1093/neuonc/noq046.PubMed CentralPubMedGoogle Scholar
- Varenika V, Dickinson P, Bringas J, LeCouteur R, Higgins R, Park J, Fiandaca M, Berger M, Sampson J, Bankiewicz K: Detection of infusate leakage in the brain using real-time imaging of convection-enhanced delivery. J Neurosurg. 2008, 109 (5): 874-880. 10.3171/JNS/2008/109/11/0874.PubMed CentralPubMedGoogle Scholar
- Holloway KL, Gaede SE, Starr PA, Rosenow JM, Ramakrishnan V, Henderson JM: Frameless stereotaxy using bone fiducial markers for deep brain stimulation. J Neurosurg. 2005, 103 (3): 404-413. 10.3171/jns.2005.103.3.0404.PubMedGoogle Scholar
- Frey S, Comeau R, Hynes B, Mackey S, Petrides M: Frameless stereotaxy in the nonhuman primate. Neuroimage. 2004, 23 (3): 1226-1234. 10.1016/j.neuroimage.2004.07.001.PubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.